The ‘s state of the art design uses DSP technology to simulate non-linear loads using a unique CZ operation mode allowing realistic loading behavior. According to the help, they should not be non-zero; that is all. Gauss fit ignores or misinterprets weights? As show, the Vpeak value is significantly reduced and incorrect when using non-synchronous loading. The also offers a unique dynamic frequency sweep with variable frequencies up to 50kHz. Low voltage operation is possible towards zero volts with corresponding reduced current levels see de-rating curves. In addition, series also provides voltage peak measurements during actual loading conditions.
|Date Added:||9 March 2013|
|File Size:||48.64 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Browsing All Articles Articles.
Chroma ATE 63206
According to the help, they should not be non-zero; that is all. I have a machine that runs a TestStand chroka for testing. I would like to program the outputs so that they are disactivated if a password has been inserted e. How can I do that? Below are some examples of the most popular program sequences available.
Chroma ATE Chroma 61700 Power Supply
The maximum power for a single mainframe is 2kW when five load modules are paralleled. The dynamic loading mode can simulate different loading conditions for most test requirements.
I of course ran into trouble with the very first VI that I am trying to upgrade. The results is not labvew at all. The system cannot find the file specified. If you guys can help me, please give areply.
Browse the Latest Snapshot. Low voltage operation is possible towards zero volts with corresponding reduced current levels see de-rating curves. Dedicated remote load sensors and control circuits guarantee minimum waveform distortion during dynamic loading.
I am looking for any help and or advice.
Now here is the tricky part. So I am wondering whether there is a way to change the Modbus address of the IO server programmatically after it has been initially configured? The ‘s state of the art design uses DSP technology to simulate non-linear loads using a unique CZ operation mode chrona realistic loading behavior.
I mean, I do not have to measure the frequency of the pulses ‘cos I already set it, I’d rather read the signal from the meter in a lavview period of time and count the number of pulses in that time. Feature is excellent for determining the peak current which can be sustained by converters.
Is it possible and how? Well, I’ve got a Gauss curve that is not so good “Gauss1” — red curve. Real time measurement of voltage pabview current are integrated into each load module using a bit measurement circuit with three current ranges.
Chroma ATE Inc.-Drivers Download
Why did Gauss Peak Fit ignored my weights? Articles on this Page showing articles to of The series load modules operate in constant voltage, current, resistance, power, or impedance to satisfy a wide range of test requirements.
I would hope not the later, as this would mean the integral function is dependent on the software loop time, and has much lesser resolution than if it uses all the data.
Gauss fit ignores or misinterprets weights? I have always used the same Modbus address to configure the IO server and there was never a need to change it until now. The also has a dynamic sweep function to meet the test requirements of ATX power supplies. Each load module contains 3 load current ranges with a minimum full current operating voltage of 0.
I have to read the signal from that instrument but I don’t really understand how. Are USB webcams compatible with labview? The 66300 electronic load mainframe accepts the user-installable series load modules for easy system configuration. When I run the vi with the highlight chhroma on, I can see both x and y data points are going into the xy graph for each change of x input.
I am using hardware timing on a cDAQ to measure analog data, and plotting the results on a waveform chart.